During equilibration and rinsing a column, the following question often arises: “For how long do I have to flush the column?” If the (re)equilibration time is too short, there is a risk of non-reproducible results (e.g. retention time shift). If in doubt a longer duration of flushing should be favoured. But if the rinse time is too long, time and, especially for preparative methods, solvents are wasted.
A YMC-Triart C8 column (TO12S03-1546WT, column volume 2.5 mL) was rinsed with 50 mL of acetonitrile before each run. A plant extract was injected under four different conditions:
- without equilibration
- equilibration with 1 column volume
- equilibration with 5 column volumes
- equilibration with 10 column volumes
It is obvious, without sufficient equilibration the retention time shifts to shorter times. When using 5 or 10 column volumes, the results of this application are reproducible. Equilibration with 5 column volumes is most efficient.
For detailed and further information, e. g. on how to calculate the column volume, download the technical document here.
Πρέπει να έχετε συνδεθεί για να σχολιάσετε.